The contains four separate circuit boards: Sign In or Register. This push-button safety system prevents the leadscrew from being engaged unless the top of the syringe is correctly positioned in the plunger clamp. No additional import charges on delivery. Trademark Information is a registered trademark of. The following display will appear and the hours digits will have a flashing highlight:
|Date Added:||19 December 2009|
|File Size:||66.52 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Potentiometer RV1 is set as follows: Only qualified personnel should maintain and repair a pump. One full turn of the adjusting nut gives approximately 2.
Oil and Coolant Circulating Heating System. Disconnect the flexible loom connector PL4 from the Main board. A super nut assembly engages onto the leadscrew, and the assembly is also connected to a steel tube.
PCA Pump Graseby 3300
Fit a new transformer by reversing steps 2 to 4 detailed above and then close the casing see on page For additional information, see the Global Shipping Programme terms and conditions – opens in a new window or vraseby.
This is used to supply high current components such as the motor thus avoiding overloading the regulator. Internally generated reset pulses will not affect the Main processor. May not post to United States – Read item description or contact seller for postage options. With the AC supply connected, up to 14 hours will be required to fully recharge low voltage batteries.
Graseby Syringe Driver with PCA available to buy online at Williams Medical Supplies
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to Watch list. Items are sold as-is with no returns or refunds available unless explicitly stated. Review pump functionality, sets and accessories Understand key process steps for April Goals Review pump functionality, sets and accessories Understand key process steps for Starting an infusion Stopping an infusion Changing a container Re-programming the pump Troubleshooting More information.
All programmed values are retained following power off or AC failure. If the appropriate pulses are not detected by the opto-sensor then an alarm signal is generated.
Graseby 3300 PCA Syringe Pump
Dartford Plug in slots K More information. A More information.
The repair procedures detailed in the sections that follow, must only be carried out by qualified personnel. Move the two blank connectors so that they cover the two unused alternative voltage terminals. The alternative settings are as follows: Infusion may be short-term up to 96 hrs.
Introduction No more bulky batteries, syginge one sleek unit More information. Redeem your points Conditions for uk nectar points – opens in a new window or tab. The battery recharging time when a pump is connected to the mains and running can be much longer. Issue 7 August Service Manual 2 9. The software for the pump uses the latest structured design methods.
The main processor must supply a command every 40 ms. The wet-side occlusion pressure monitoring made the pump particularly suitable for use in intensive-care baby units.
Failure to use the mains lead clamp means that the pump may be accidentally or sydinge disconnected from the mains.
The thrust of a particular pump may, therefore, differ from the original factory set level. Parameters and options Table 2. If the pump is used in the presence of, or in combination with Magnetic Resonance Imaging MRI machines it must be protected from the magnetic field emitted by such equipment.
Used GRASEBY programable Pump PCA For Sale – DOTmed Listing #
This flag rotates about the axis of the hinge and operates an opto-sensor on the Main board which informs the processor whether the cover is open or closed. Temporarily assemble the halves of the pump, being careful not to trap any leads.
Trademark Information is a registered trademark of.